Welcome To Our Store!
Add Your Review
Checked products are added to the cart with this item. select all
Available in 4 standard unisex sizes, to select your size please consider the following:
Please Note: the options selected above determine the final part number for this product. If you are submitting information to your purchasing department to purchase this item on a PO, please add the item to your cart with your desired options selected and click "proceed to checkout" The accurate/complete product code & price are listed in the My Shopping Cart section at the bottom of the screen.
Only registered users can write reviews. Please, log in or register
Question will be displayed after moderation.
All answers will be displayed after moderation.
Warranty provides coverage due to defective parts or workmanship for a period of two (2) years after the date of original purchase. This warranty does not cover damages due to misuse or abuse of the product nor damages related to uses outside the designed parameters of the unit.
With careful use and proper storage, the useful life of the garment should be several years. When not in use, garments should be placed on a hanger. Extreme heat will severely damage your garment. Avoid contact with hot surfaces or exposure to direct sunlight. Also avoid contact with sharp objects, as these may penetrate the garment's protective core. For transportation, the garments should be carefully rolled up and put in a box or tube.
Many aqueous-based stains, especially from body fluids, are easily removed by wiping with a cloth/sponge dampened with water containing a non-aggressive, non-bleach detergent. A soft scrubbing brush may be needed for dried blood and stubborn aqueous stains (including barium contrast media) where these have gone into the weave/fibers of the fabric.
DO NOT IRON - DO NOT DRY-CLEAN - DO NOT AUTOCLAVE - DO NOT USE BLEACH - DO NOT USE ALKALI DETERGENTS - DO NOT USE ALCOHOL
Regular inspections of the garment, at least annually, are recommended using radiographic or fluoro operating at 70-90 kV. Lines may indicate cracks; dots may be pinholes, (seen as dark on film, clear on fluoro) - consult your Radiation Safety Officer.
Add Your Review
* Required Fields