IBI Quick Screen Horizontal Electrophoresis System is constructed of 3/8" thick durable cast acrylic that resists cracking and warping for years of use.
Designed for rapid linear separations of nucleic acids. It requires minimal setup and provides excellent band resolution with consistent lane-to-lane mobility for accurate results. It is versatile for use with applications requiring quick screens, such as fragment analysis and recombinant screening, all in 30 minutes or less.
The maximum suggested applied voltage for the electrophoresis of DNA in agarose gels, using the QS-710, is 100 volts. In a 1% TBE gel, this translates into a run time of approximately one hour. Lower voltages may be used with the general rule that a 50 V run will take twice as long as a 100 V run. Higher voltages may be used to decrease the run time; however, the heat generated during high voltage electrophoresis may decrease sample resolution. Such artifacts may be avoided by running the unit in a cold room or by adding 1 X electrophoresis buffer "ice cubes" to keep the unit properly cooled.
The suggested run parameters for the electrophoresis of RNA in agarose gels containing formaldehyde is 60-80 volts. Since the QS-710 has no capacity for buffer recirculation, the buffer from each reservoir should be collected after 1.5 hours of electrophoresis.
Casting Tray bottom is constructed of UV Transmittance cast acrylic to allow viewing of gel with a UV light source.
The IB51000 has a patented vented lid dissipates heat for cool and fast separations. Patent# 5,779,869.
The Unit Comes Complete With:
Model QS-710 Buffer Tank
Patented QS-710 Vented Lid
UV Transmittant Casting Tray (IB51020)
Casting Fixture (IB51025)
Two 1.5mm x 8 Tooth Combs
One Set of Power Cords (Red and Black)
Unit Dimensions: 6.4cm (H) x 11.4cm (W) x 22.2cm (L)
Gel Dimensions: 7cm (W) x 10cm (L)
Maximum Sample Capacity: 30 Samples (2 Combs x 15 Samples Each)
Warranty: IBI's limited warranty for all electrophoresis gel boxes is four (4) years to the original buyer only (nontransferable). Warranty does not apply to electrodes or platinum wires.
If you would like to return an item, please contact customer service within 30 days of product shipment to receive a return authorization (RA) number. All returns must include the RA number, include original packaging and be shipped pre-paid. Items returned without proper authorization will not receive credit. A 15-25% restocking charge may be assessed for returned items. All shipping charges remain payable.